Visit us on the web at for more information: • Price Quotations • Drivers· Technical Specifications. Manuals and Documentation . Model System DMM Instruction Manual Rev. D. Manual Type: Primary User; Part Number: D. Release Date: Sep 17 View and Download Keithley DMM instruction manual online. 5 function autoranging digital multimeter. DMM Measuring Instruments pdf manual.
|Published (Last):||16 February 2014|
|PDF File Size:||5.51 Mb|
|ePub File Size:||12.63 Mb|
|Price:||Free* [*Free Regsitration Required]|
Enter the number 6 by pressing the “6” button.
Each press of one of these buttons toggles the display between the two available conditions. For high impedance sources, the generated noise 19 become significant when using the most sensitive range 3WmV, 6V2d of the Model The leads to the input may be shielded or unshielded, as necessary.
If the filter value is 5, the follow- ing message wUl be displayed: The most recent defined word will be listed first. Perform the following general procedure to trigger readings into data store from the front panel: Manul help on what type of coding to use would be greatly appreciated.
Copyri 67 68 74 20 28 63 29 20 31 39 39 34 41 6C 6C 20 72 69 67 68 74 73 20 72 65 73 65 72 76 65 64 2E ght c All rights reserved. The follow- ing paragraphs indicate factors which affect accuracy, noise, source resistance, thermal emfs and stray pick-up.
Keithley 196 DMM Instruction Manual
Table lists and briefly describes the available front panel programs. In this instance, the following error message will be displayed on the front panel: Therefore, press a Range button and the following message will be displayed: A consideration of filter usage is that the larger the weighting, the longer the response time of the display.
To change to a different measuring function, simply press the desired function button. keitbley
Video AVI1. March 16,1: The Translator buffer can hold approximately character Translator words.
System DMM Instruction Manual | Keithley – TechRecovery, LLC
Device-dependent commands can be sent either one at a time, or in groups of several commands within a single string. To return the instrument to the factory power up default conditions, use Program 37 Reset and save the condi- tions using Program The following two examples will use the maximum allowable zero values counts and counts to show that dynamic measurement range will not be reduced. The instrument will then respond to the Translator words as well as Keithley device-dependent commands.
The instrument will return to the previously defined state.
When the B1 command is sent, subsequent keituley will be taken fram consecutive buffer locations beginning with the first memory location The reading is sensitive to and responds to temperature changes. Enter the number 5 by pressing the “5” button. The reading mode commands are as follows: Often the user may incorrectly suspect that the hardware is causing a problem, when it was the software all along.
For exam- ple, if the LO limit is the factory default value, the majual ing message will be displayed: However, each device on the bus must have a unique primary address– a factor that should be kept in mind when setting keirhley primary address of the Model Keithlsy any of these commands are transmitted, the instrument will send the appropriate reading the next time the instrument is addressed to talk. Source Measure Unit Instruments. A clean copper conductor such as 10 bus wire is about the best for this application.
Connect the test leads to the input of the Model and short them together. December 5, at 9: Perform the following steps to run this program: If ohms offset compensation was disabled, the instru- ment will return to the previous operating state.
Set the Model to ACV and autorange. Exploring the boundaries of materials science or device development?